


Detector type  Optical specifications  Mechanical specs 



Part
number 
Mag.  Image
rectangle 
1/3”
6.0 mm diag w × h 4.80 x 3.60 
1/2.5”
7.1 mm diag w × h 5.70 x 4.28 
1/2”
8.0 mm diag w × h 6.40 x 4.80 
1/1.8”
8.9 mm diag w × h 7.13 x 5.33 
2/3”  5 MP
11.1 mm diag w × h 8.50 x 7.09 
WD  wF/#  Telecentricity
(max) 
Distortion
(max) 
Residual
Distortion 
Field
depth 
CTF @
70 lp/mm 
Mount  Phase
adj. 
Dimensions 



(x)  (mm)  (mm × mm)  (mm × mm)  (mm × mm)  (mm × mm)  (mm × mm)  (mm) 

(deg)  (%) 

(mm)  (%) 


(mm)  (mm)  (mm) 



1 





2  3  4  5  6  7 


8  9  10  



Object field of view (mm x mm) 









A  B  C 


TCCP12144Ultra compact bitelecentric lens for matrix detectors up to 1/1.8”, magnification 0.044x  0.044  7.48 x 5.60  109.1 x 81.8  129.5 x 97.3  145.4 x 109.1  162.0 x 121.1  217.4  8  < 0.06 (0.1)  < 0.8  < 0.01  214  > 45  C  Yes  332.0  302.5  299.4  Buy now  
TCCP23144Ultra compact bitelecentric lens for matrix detectors up to 2/3”, magnification 0.059x  0.059  8.93 x 7.45  81.9 x 61.4  97.3 x 73.0  109.2 x 81.9  121.7 x 91.0  145.1 x 121.0  217.4  8  < 0.06 (0.1)  < 0.8  < 0.01  121  > 45  C  Yes  332.0  302.5  315.1  Buy now 
TCCP12192Ultra compact bitelecentric lens for matrix detectors up to 1/1.8”, magnification 0.033x  0.033  7.48 x 5.60  145.9 x 109.4  173.3 x 130.1  194.5 x 145.9  216.7 x 162.0  328.0  8  < 0.12 (0.18)  < 0.8  < 0.01  382  > 45  C  Yes  410.4  344.1  345.0  Buy now  
TCCP23192Ultra compact bitelecentric lens for matrix detectors up to 2/3”, magnification 0.046x  0.044  8.93 x 7.45  109.6 x 82.2  130.1 x 97.7  146.1 x 109.6  162.8 x 121.7  194.1 x 161.9  328.0  8  < 0.12 (0.18)  < 0.8  < 0.01  216  > 45  C  Yes  410.4  344.1  353.3  Buy now 
TCCP12260Ultra compact bitelecentric lens for matrix detectors up to 1/1.8”, magnification 0.027x  0.027  7.48 x 5.60  177.8 x 133.3  211.1 x 158.5  237.0 x 177.8  264.1 x 197.4  366.0  8  < 0.18 (0.22)  < 0.9  < 0.01  568  > 45  C  Yes  425.3  396.7  421.0  Buy now  
TCCP23260Ultra compact bitelecentric lens for matrix detectors up to 2/3”, magnification 0.036x  0.036  8.93 x 7.45  133.3 x 100.0  158.3 x 118.9  177.8 x 133.3  198.1 x 148.1  236.1 x 196.9  366.0  8  < 0.18 (0.22)  < 0.9  < 0.01  568  > 45  C  Yes  425.3  396.7  421.0  Buy now 