Detector type | Optical specifications | Mechanical specifications | |||||||||||||||||
---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|

Part number | Mag. | Image circle Ø | 1/3″ 6.0 mm diag w x h (4.80 x 3.60) | 1/2.5″ 7.1 mm diag w x h (5.70 x 4.28) | 1/2″ 8.0 mm diag w x h (6.40 x 4.80) | 1/1.8″ 8.9 mm diag w x h (7.13 x 5.33) | 2/3″ 11.1 mm diag w x h (8.50 x 7.09) | WD | wF/# | Telecentricity | Dist. | Field depth | CTF @ 70 lp/mm | Mount | Phase adj. | Length | Max height | Max width | |

(x) | (mm) | (mm x mm) | (mm x mm) | (mm x mm) | (mm x mm) | (mm x mm) | (mm) | (deg) | (%) | (mm) | (%) | (mm) | (mm) | (mm) | |||||

1 | 2 | 3 | 4 | ||||||||||||||||

Object field of view (mm x mm) | |||||||||||||||||||

TCZR036SBi-telecentric zoom with motorized controls, magnification 0.25x to 2x, for detectors up to 2/3″ | 0.250 0.500 1.000 2.000 | 11.0 | 19.2 x 14.4 9.6 x 7.2 4.8 x 3.6 2.4 x 1.8 | 22.8 x 17.1 11.4 x 8.56 5.70 x 4.28 2.85 x 2.14 | 25.6 x 19.2 12.8 x 9.6 6.4 x 4.8 3.2 x 2.4 | 28.5 x 21.3 14.3 x 10.7 7.13 x 5.33 3.57 x 2.67 | 34.0 x 28.4 17.0 x 14.2 8.50 x 7.09 4.25 x 3.55 | 74.0 | 16 | < 0.05 | < 0.05 < 0.04 < 0.04 < 0.08 | 13.2 3.3 0.8 0.2 | > 40 > 35 > 40 > 35 | C | Yes | 212.0 | 144 | 103 | Buy now |

TCZR072SBi-telecentric zoom with motorized controls, magnification 0.125x to 1x, for detectors up to 2/3″ | 0.125 0.250 0.500 1.000 | 11.0 | 38.4 x 28.8 19.2 x 14.4 9.6 x 7.2 4.8 x 3.6 | 45.6 x 34.2 22.8 x 17.1 11.4 x 8.56 5.70 x 4.28 | 51.2 x 38.4 25.6 x 19.2 12.8 x 9.6 6.4 x 4.8 | 57.0 x 42.6 28.5 x 21.3 14.3 x 10.7 7.13 x 5.33 | 68.0 x 56.7 34.0 x 28.4 17.0 x 14.2 8.50 x 7.09 | 157.8 | 16 | < 0.05 | < 0.1 < 0.08 < 0.05 < 0.07 | 53.0 13.2 3.3 0.8 | > 35 > 40 > 40 > 35 | C | Yes | 279.7 | 144 | 103 | Buy now |